Here we have provided comprehensive lists of commonly asked questions regarding our fluidics products and related applications. This information is designed to support your inquiries, but if you don’t find the answers you are looking for we encourage you to contact us for further assistance.
Please use the sort buttons in the left navigation to navigate between questions and answers that are specific to our Optical Filters, Optical Imaging Systems, Cameras, Microfluidics, Webstore, and Engineering Partnership.
An injection valve serves two purposes. First, it allows the redirection of the pump flow away from the sample holding loop, giving you the opportunity to fill the loop with a predetermined amount of sample. Second, it once again allows the redirection of the pump flow through the sample loop, where the solvent stream carries the sample slug onto the head of the column to begin the separation process.