Here we have provided comprehensive lists of commonly asked questions regarding our fluidics products and related applications. This information is designed to support your inquiries, but if you don’t find the answers you are looking for we encourage you to contact us for further assistance.
Please use the sort buttons in the left navigation to navigate between questions and answers that are specific to our Optical Filters, Cameras, Microfluidics, Webstore, and Engineering Partnership.
First, confirm that the EV/MXII valve will operate separately from your instrument. To isolate the valve, disconnect the interface cable from your instrument, leave the power supply connected to the unit, and put the unit in the LOCAL mode using the front panel.
To check the communications in the level logic mode:
To check the communications in the BCD mode:
If the valve is responding as described, the problem is with the instrument’s communication to the EV/MXII unit. Contact your instrument manufacturer’s technical service department.