Here we have provided comprehensive lists of commonly asked questions regarding our scientific cameras. This information is designed to support your inquiries, but if you don’t find the answers you are looking for we encourage you to contact us for further assistance.
Please use the sort buttons in the left navigation to navigate between questions and answers that are specific to Fluidics, Optical Filters, Microfluidics, our Webstore and Engineering Partnership.
IDEX Health & Science uses Peltier devices, which are thermoelectric coolers that get cold on one side and hot on the other when electricity is applied. Designing a CCD chamber than can keep a seal for many years is part of the engineering; efficiently dissipating the heat generated by the cooling is another part. Typically the temperature sensor for a camera is installed in a copper block that links the CCD to the Peltier cooler. If there is a problem with contact between the copper block and the sensor, the block may be cold but the sensor may not be fully cooled. In this case, you can see dark current values that are inconsistent with reported temperature of the camera.